Any dry bulk materials: ferroalloys, ores, granite, soil, sand, coal, glass.
|Maximum material feed size, mm
|Riffle width, mm
|Number of riffles
|Gross/usable capacity of the sample collector, l
|Overall dimensions with 2 sample collectors (Length x Width x Height), mm
|Overall dimensions of 3rd sample collector (Length x Width x Height), mm
|Weight, including 3rd sample collector, kg
|Material of splitting unit and sample collectors
Sample division error does not exceed 2%.
NO CROSS-CONTAMINATION OF SAMPLES
The splitting unit and sample containers are made of 08X18H10T stainless steel (AISI 304).
Small dimensions and weight.
The detachable design of RSS 5 provides easy cleaning of residual material. The sample splitter is equipped with three sample collectors, each of them can be used as a loading scoop. That allows to repeat the division of the sample many times.
DESCRIPTION OF THE DEVICE
The structure of the riffle sample splitter includes: a splitting unit, sidewalls, racks, three sample collectors.
The splitting unit is a package of plates forming riffles directed alternately to the sides. The block is covered by casing with loading funnel in the upper part.
Samplers are designed to collect a divided sample for uniform loading of the sample over the entire width of the loading funnel.
The riffle sample splitter must be placed horizontally on a flat surface. Samplers should be inserted under the splitting unit until they stop. The sample is fed to a loading funnel. The material enters the splitting unit and is evenly distributed along the riffles to the sample collectors. Parts of the sample are transferred for further processing or re-division.
RSS 5 riffle sample splitter on T 70 stand