Maliy pr. V. O., 62/2, liter A, St. Petersburg, Russian Federation, 199178
Any dry bulk materials: ferroalloys, ores, granite, soil, sand, coal, glass.
|Maximum material feed size, mm||15|
|Riffle width, mm||25|
|Number of riffles||20|
|Gross/usable capacity of the sample collector, l||17,5/ 12,5|
|Overall dimensions with 2 sample collectors (Length x Width x Height), mm||600х546х403|
|Overall dimensions of 3rd sample collector (Length x Width x Height), mm||510х270х261|
|Weight, including 3rd sample collector, kg||27,5|
|Material of splitting unit and sample collectors||AISI 304|
Sample division error does not exceed 2%.
NO CROSS-CONTAMINATION OF SAMPLES
The splitting unit and sample containers are made of 08X18H10T stainless steel (AISI 304).
Small dimensions and weight.
The detachable design of RSS 25 provides easy cleaning of residual material. The sample splitter is equipped with three sample collectors, each of them can be used as a loading scoop. That allows to repeat the division of the sample many times.
DESCRIPTION OF THE DEVICE
The structure of the riffle sample splitter includes: a splitting unit, sidewalls, racks, three sample collectors.
The splitting unit is a package of plates forming riffles directed alternately to the sides. The block is covered by casing with loading funnel in the upper part.
Samplers are designed to collect a divided sample for uniform loading of the sample over the entire width of the loading funnel.
The riffle sample splitter must be placed horizontally on a flat surface. Samplers should be inserted under the splitting unit until they stop. The sample is fed to a loading funnel. The material enters the splitting unit and is evenly distributed along the riffles to the sample collectors. Parts of the sample are transferred for further processing or re-division.
T 70 stand